Description of the payload
An instrument optimized for the measurement of the cosmic ray antiproton flux up to at least a range of energy of 50 GeV, that identifies particles through the combination of a superconducting magnet, a drift tube hodoscope (DTH), time-of-flight scintillators (ToF), and multiple energy loss wire chambers (dE/dx). At left can be seen a scheme of it (click to enlarge)
Its an evolved version of a similar experiment that was focused on the detection and identification of cosmic-ray electrons and positrons, flown succesfully in 1994 and 1995.
The superconducting magnet, with two circular Niobium-Titanium coils held at 4.2 K in a liquid-helium bath, provides an approximately uniform field of about 1 Tesla within a rectangular room-temperature bore. The cryostat employs a vapor-cooled radiation shield and superinsulation to isolate the inner helium vessel from the ~300 K environment. The total liquid helium capacity for the system is 260 liters, with a maximum hold time of ~7 days.
The drift-tube hodoscope (DTH) consists of 479 thin-walled drift tubes, each 2.5 cm in diameter, arrayed in 26 layers. Of these, 18 layers are aligned with the tube axes parallel to the magnetic field, and provide trajectory measurements used to determine rigidity. The remaining 8 layers, aligned with axes normal to the magnetic field, provide the additional information required to reconstruct trajectories in three dimensions. A drift gas based on CO2 with hexane as an organic quencher flows through the tubes.
The time-of-f1ight detector discriminates against upward particles, which can masquerade as antiprotons. In addition, ToF signals are combined with coincident hits in S (a separate layer of scintillator in the magnet bore just above the DT hodoscope) to provide a fast trigger. The ToF detector consists of two 1 m x 1 m layers of plastic scintillator separated by 2.76 m. Each scintillator layer is divided into four paddles 1 cm x 25 cm x 1 m in size and each end of a paddle is viewed by a photomultiplier tube (PMT). Each PMT detects approximately 40 photoelectrons for a vertical atmospheric muon traversing the center of the paddle. For a triggered event, timing and charge-integrated signal size are digitized for each participating PMT. Thus, upward and downward going particles are easily distinguished by the TOF detector.
The dE/dx detector system contains a total of 140 multiwire proportional chambers, each with a thickness of 1 cm, and is divided into two stacks, one above and one below the magnet. A combination of Xenon and Methane (95% and 5% of the mixture respectivelly) provides the highest possible velocity resolution and stability. Anode wires are connected in groups of 4-6 and read out by an AMPLEX amplifier. Custom 8-bit ADCs digitize the amplified signals.
Details of the balloon flight and scientific outcome
Launch site: Scientific Flight Balloon Facility, New Mexico, US
Balloon launched by: National Scientific Balloon Facility (NSBF)
Balloon manufacturer/size/composition: Zero Pressure Balloon 39.570.000 cuft - SF3-39.57-.8/.8/.8/.8-NHR
Balloon serial number: W39.57-3-16
Flight identification number: 465N
Useless data due to a malfunction in the voltage level of the instrument. Fixed for future flights
External references and bibliographical sources