Description of the payload
HIREGS was a high resolution gamma-ray spectrometer, developed as part of NASA Space Physics Division's MAX 91 Program to study solar flare activity from long duration balloons during solar cycle 22. The study of flare gamma-ray emissions was aimed to obtain a better understanding of the relation of solar flares to the larger picture of solar activity, the solar cycle, and the field/particle environment of the earth.
The HIREGS instrument consisted of an array of twelve 6.7 cm diameter x 6.2 cm long n-type germanium (Ge) closed end coaxial detectors which were cooled by three 50-liter liquid nitrogen (LN) dewars. Eight of the detectors were electrically separated into front and rear segments to improve the low energy background and flare performance. The detector array was surrounded by an active shield that reduced the gamma-ray background by allowing the anticoincidence of simultaneous detector and shield events. The shield system consisted of 5 cm thick Bismuth germanium oxide (BGO) segments at the rear and sides and a 10 cm thick drilled Caesium iodide (CsI) collimator on the front which determined a field of view of 24º. Additional hard X-ray (<200 KeV) passive collimators inserted in the CsI collimator covered half of the detectors, limiting these fields of view to 3.7º x 24º full width at half maximum (FWHM), allowing separation of point source spectra from the diffuse Galactic continnum. The detectors, LN dewars, shield, and collimator were held in a cradle structure that was in turn supported within the balloon gondola.
The gondola frame was 7 x 9 ft. wide x 7 ft. tall. Including the solar panels, it spanned 17 ft., and including the tripod and rotor assembly, above, and the SIP solar panels, below, it was -19 ft. tall.
All pointing was determined relative to the Sun and was controlled by the Gondola Control Unit (GCU). The GCU was an independent computer dedicated to pointing operations, the collection of housekeeping data, and the acquisition of GPS information. The GCU also controled various power relays and could receive commands from the main computer as well as from the several telemetry command links. The detector craddle could be pointed for solar and celestial observations with an accuracy of 0.5º both in elevation and azimuth. The Azimuth orientation of the instrument was determined by the relative position of the azimuth Sun sensors with respect to the gondola. When the gondola was pointed to the Sun in Azimuth, the elevation drive could be operated in a feedback loop with Sun sensors mounted on the cradle axis to locate the Sun in elevation.
Power was provided by a 8.1 m2 solar array made up of cristalline silicon photo-cells, arranged on two lateral panels that delivered up to 1000 watts of power during the flight. Also, a set of pressurized lead-acid batteries located in the electronics bay could power the instrument for several hours if the solar panels were pointed away from the Sun.
A computer controled the instrument commanding and housekeeping, and also packed the data and recorded it in an Exabyte tape storage system which could save up to 4.6 Gbytes of information (enough to hold all the data generated in a 3 week balloon flight). Also the data was fed to the telemetry system which could send it to a ground station or to an LC-130 aircraft that was sent to underflight the balloon every few days. That system served as a backup in case the onboard storage tape could not be recovered after the flight.
HIREGS counted with two communication systems, a Line-Of-Sight (LOS) telemetry available only when the balloon was within the range of a ground station or a underflying aircraft and a high frequency (HF) link which could be used during the entire flight to send commands to the instrument. Complementary, ARGOS and INMARSAT satellite based systems were also available.
Modification from previous flights
For this fourth and last flight of HIREGS, one of its Germanium detectors was replaced by a narrow-bore coaxial Germanium gamma-ray detector, and electronics of a prototype developed for the pulse shape discrimination (PSD) system to be installed onboard INTEGRAL/SPI. The first goal for this test flight was to demonstrate the performance of the PSD electronics in a radiation background and space environment similar to that in which SPI will operate. The second goal was to measure the fraction of the background due to localized beta decays, and to validate the model used to predict SPI's background and the sensitivity improvement due to the PSD background rejection.
Details of the balloon flight and scientific outcome
Launch site: Williams Field, McMurdo Station, Antarctica
Balloon launched by: National Scientific Balloon Facility (NSBF)
Balloon manufacturer/size/composition: Zero Pressure Balloon 29.470.000 cuft (SF3-424.37-080-NSC-X-ST)
Balloon serial number: W29.47-2X-14
Flight identification number: 444N
Payload weight: 3130 lbs
Overall weight: 4809 lbs
The instrument was launched by dynamic method using a delta vehicle fitted with a crane on 7 January, 1998 from Williams Field, McMurdo Station, Antarctica. The balloon reached its maximum altitude of 38 km a few hours after launch and started the typical anticyclonic path as expected. 48 hours into the flight, instead of maintain the float altitude, it started to descend steadily with no reason. When the NSBF team moticed this, they sent commands to the balloon to release ballast in a series of efforts to maintain altitude, to no avail. Thus, on 11 January, they decided to terminate the flight. The payload landed safely near Vostok Station and automatic geophysical observatory 4 (AGO-4). A few hours later, the Air National Guard flew an LC-130 airplane to the AGO-4 site and recovered the balloon's payload.
The shortened flight and low altitudes attained due to the leak experienced by the balloon compromised HIREGS's scientific goals, which obtained no useful data. However, on regard the testing of the PSD system, the mission was a complete success, obtaining very useful data for the future of the project.
External references and bibliographical sources
- HIREGS website Space Physics Research Group - University of California, Berkeley
- A high-resolution gamma-ray and hard X-ray spectrometer for solar flare observations in Max 1991 Max 1991: Flare Research at the Next Solar Maximum. Workshop 1: Scientific Objectives; p 262-278
- Balloon Flight Terminated Antarctic Sun - 24/1/1998 (PDF Format)
- Balloon flight test of pulse shape discrimination (PSD) electronics and background model performance on the HIREGS payload Nuclear Instruments and Methods in Physics Research Section A: Accelerators, Spectrometers, Detectors and Associated Equipment - Vol. 491, Issue 3, Pag.390
- Ballooning Over Antarctica - Science at 125,000 feet Antarctic Sun - 10/1/1998 (PDF Format)
- Gamma-ray spectroscopy: Galactic center observations and germanium detector development Thesis (PHD) by Steven E. Boggs, UCB, Source DAI-B 59/08, p. 4171, Feb 1999, 160 pages