The Super-pressure Balloon-borne Imaging Telescope, also known as SuperBIT, is an astronomical instrument designed to demonstrate the capabilities and performance of high-gain stability, on-demand, wide-field, visible-to-near-UV imaging from the stratosphere hanging from a Super Pressure Balloon (SPB) aimed to perform strong/weak lensing experiments, exo-planetary research, and other cosmological studies. Operating in the visible-to-near-UV bands of the spectrum, made SuperBIT a very unusual balloon-borne telescope which usually operate in the Infrared, Gamma or Submilimeter domains. The last balloon-borne instrument that performed observations in the visible was the infamous and gargantuan Stratoscope-II which flew a dozen of times in the 60's and early 70's.
The instrument houses a 0.5 meter Ritchey-Cretien-type telescope with refractive field correction optics providing a relatively wide 0.5º field-of-view, which is stabilized in all three rotational axes. At left we can see an scheme of the instrument (click to enlarge). Mechanically, SuperBIT is a three-axis stabilizer that compensates for the rough compound pendulative effects that manifest in the balloon-borne environment while compensating for field rotation effects over a relatively wide 0.5º field-of-view. When fully assembled, the SuperBIT instrument is approximately 3 m tall from the base to the flight train connection or pivot and weighs approximately 1000 kg, which is within the weight class suitable for SPB missions. The three mechanical degrees-of-freedom are constructed from a series of nested frames: an outer frame, which controls gondola yaw; a middle frame, which actuates gondola roll about the outer frame; and an inner frame, which controls the telescope pitch about middle frame. Constructed primarily from aluminum honeycomb panels, these three nested frames are both lightweight and structurally stiff, which reduces the potential for the driving of structural modes that can degrade pointing performance and subsequent image quality. While the full 360º yaw range is accessible from altitude, the roll of the middle frame is constrained to ±6º due to inter-frame interference and the pitch of the inner frame is restricted at 20º on the lower end due to the Earth's limb and 55º on the upper end from the angular obstruction caused by the balloon at full expansion.
In order to provide diffraction limited imaging at the telescope focal plane with a 0.02" effective image stability, SuperBIT architecturally refines its pointing precision through three successive pointing and stabilization regimes. Firstly, targets are acquired for accessible regions on the sky by slewing the three gimbal frames to coordinates corresponding to the desired right ascension (RA) and declination (Dec) of the target. Gondola yaw is coarsely achieved via a large reaction wheel, which is driven by a three-phase torque-controlled frameless motor on the base of the outer frame, with inertial and absolute feedback provided by fiber optic rate gyroscopes and a three-axis magnetometer. To prevent reaction wheel saturation due to persistent disturbances, excess angular momentum is dumped through the gondola and along the flight train to the balloon via a speed-controlled two-phase pivot stepper motor, which provides the necessary gondola-external torque by manipulating flight train twist. Along the roll and pitch axes, the middle and inner frames, respectively, are each actuated about their respective frames via pairs of high-inductance three-phase frameless motors equipped with flexure bearings, to prevent non-linear static friction effects, as well as high resolution absolute optical encoders for relative gimbal position feedback. In order to access the full elevation range of the telescope, the pitch axis of the inner frame is co-actuated by a pair of two-phase stepper motors through a high resolution 100:1 gear box, which together coarsely acquires the necessary pitch gimbal position to within < 1'.
After coarse target acquisition, a pair of orthogonal star cameras provide both absolute pointing information through so-called lost-in-space routines using the pattern recognition routines developed for astrometry as well as differential feedback for high bandwidth control telescope stabilization at the sub-arcsecond level through high resolution centroiding. The star cameras can centroid to sub-arcsecond accuracies, and are sensitive to stars of at least Mag 9, which gives on average 4 stars per field of view. Although lost-in-space solutions provide a full pointing solution every 2 seconds, higher rate differential feedback is necessary to mitigate mid-frequency perturbations in the balloon-borne environment. In this differential mode, the star cameras provide star centroiding of the brightest stars in their fields-of-view in order provide frameless motor feedback with the corrections necessary to maintain telescope stability whilst serving as an absolute reference to continually correct for drifting fiber-optic rate gyroscope biases. With the coarse stepper motors locked during fine telescope stabilization, the combination of rate gyroscopes and star cameras provide feedback for frameless motors on all three gimbaled frames to provide the necessary three-axis stability for the telescope.
Once stabilized to < 1", the telescope back-end optics further reduce the focal plane disturbances to 0.02" using a piezo-actuated tip/tilt fold mirror along the optical path of the telescope between the corrective lenses and the science camera CCD. In order to provide feedback for the tip-tilt actuator, a portion of the science focal plane is picked-off and redirected to focal plane star camera that provides high rate differential feedback through sub-pixel centroiding. This centroid information is improved by incorporating rate gyroscope data such that the effective correction bandwidth of the image stabilizer is limited by the inertia of the fold mirror used for tip-tilt corrections as well as the relative power of the piezo-electric controller. Given the potential for thermal misalignments along the telescope bore sight, the focal plane star camera is equipped with an independent linear actuator to decouple its focus control from the telescope secondary focus position. Systematic calibrations between the telescope stabilizing star cameras and the science camera focal plane are determined through similar lost-in-space routines performed on the science camera focal plane. Lastly, in-flight telescope re-alignment is made possible via secondary mirror actuators that provide tip/tilt corrections that correct for potential optical misalignments suffered due to thermal effects or launch shocks.
Balloon launched on: 6/6/2018 at 00:16 utc
Launch site: Columbia Scientific Balloon Facility, Palestine, Texas, US
Balloon launched by: Columbia Scientific Balloon Facility (CSBF)
Balloon manufacturer/size/composition: Zero Pressure Balloon Raven - 29.000.000 cuft
Flight identification number: 1599P
End of flight (L for landing time, W for last contact, otherwise termination time): 6/6/2018 at 20:35 utc
Balloon flight duration (F: time at float only, otherwise total flight time in d:days / h:hours or m:minutes - ): 21 h 13 m
Landing site: 37 nautic miles SE of San Angelo, Texas, US
The balloon was launched by dynamic method at 20:05 utc on June 5, 2018, from the NASA balloon base in Palestine, Texas.
The balloon moved first to the southwest and two and a half hours into the flight it reached float altitude about 110.000 ft shifting course to the westerly-ward flight path that would maintain until the end. The flight progressed very well overnight, and by sunrise, with the scientific portion of the flight already completed, the balloon continued it's trip flying over central Texas while the staff of the Columbia Scientific Balloon Facility searched for a suitable place to make it land. As Superbit was floating over towns and mountains, a decision had to be taken: to make it land over rough terrain or possible military land or wait another 10 hours till it could float west to an easier place to recover it. One thing to consider was the damage that the instrument could suffer in rough terrain or how long will be the sun exposure that it could receive until the recovery party could reach it, and that could affect the optical system of the equipment.
Finally, the flight was terminated at 15:40 on June 6, with the payload landing 45 minutes later in an unpopulated zone of the Tom Green County, 37 nautic miles southeast of San Angelo, Central Texas. Total flight time was 21 hours 13 minutes.
If you consider that this website is interesting or useful, you can help to keep it running with just the equivalent of the price of a cup of coffee. Click on the button on the right for more information.