It is an uprgraded version of the RADIBAL (RADIometre BALlon) infrared polarimeter developed by LOA during 1998/1999 and used to observe aerosols ant other particles in the atmosphere.
The instrument has been miniaturized, to allow an easier launch without an auxiliary balloon. The measurements are performed in 5 channels (instead of 2 in RADIBAL) centered in the near infrared at 730 nm, 865 nm, 1000 nm, 1270 nm and 1620 nm. The three first channels are equipped with silicium detectors while the two others are InGaAs detectors cooled by a one stage Peltier module.
The instrument is composed of 15 separate optics whose axes are parallel, with narrow fields of view. They are arranged in groups of 3 for each wavelength, the 3 optics being equipped with analyzers positioned 60º one from each other in order to derive the polarized light. The radiometers detect the sunlight scattered by the atmosphere (gas and particles) enabling radiance and degree of polarization of the scattered sunlight to be inferred. Owing to the rotation of the gondola around its vertical axis, the observations are performed under various directions in a horizontal plane. Scans are thus obtained with an angular sampling of about 2-6º depending on the rotation rate.
A magnetometer helps to determine the azimuth direction of the line of sight and to deduce the corresponding scattering angle with respect to the solar incident light.
Balloon launched on: 7/7/1999 at 18:38
Launch site: Aérodrome de Gap-Tallard, Haute Alpes, France
Balloon launched by: Centre National d'Etudes Spatiales (CNES)
Balloon manufacturer/size/composition: Zero Pressure Balloon model 12sf Zodiac - 12.000 m3
Balloon serial number: 12SF Nº 104
End of flight (L for landing time, W for last contact, otherwise termination time): 7/7/1999 at ~ 21:00
Balloon flight duration (F: time at float only, otherwise total flight time in d:days / h:hours or m:minutes - ): 2 h 21 m
Payload weight: 221 kgs
Gondola weight: 24 kgs
If you consider that this website is interesting or useful, you can help to keep it running with just the equivalent of the price of a cup of coffee. Click on the button on the right for more information.