It is an advanced Fourier Transform Infra Red (FTIR) spectrometer specially tailored to the operation on a stratospheric balloon gondola. It allows precise limb emission sounding of chemical constituents related to the stratospheric ozone problem and the greenhouse effect. This method is appropriate to obtain vertical profiles of ozone and a considerable number of key radicals (NO, NO2), reservoir species (HNO3, N2O5, ClONO2, and HO2NO2) as well as source gases (CH4, N2O, H2O, CFC-11, CFC-12, CFC-22, CCl4, CF4, C2H6, and SF6) simultaneously, with an altitude resolution of 2 to 3 km.
The MIPAS-B experiment was also thought as precursor for a space version of MIPAS actually installed onboard ENVISAT.
The instrument is divided in five operational segments:
(1) the gondola,
(2) the line-of-sight (LOS) stabilization and reference system,
(3) the cryogenic spectrometer,
(4) the on-board electronics, and
(5) the ground control equipment
The gondola is a frame construction, developed by the Geneva Observatory, combining high stability and safety with relatively low weight and easy servicing.
The LOS system is based on a miniaturized inertial navigation system with embedded GPS (Global Positioning System) that provides the attitude and heading reference of the instrument frame needed for the control loop to maintain the LOS within 300 m at the tangent point. A CCD star camera takes images of stars in the direction of the LOS that are used as absolute reference.
The spectrometer consists of a three-mirror off-axis telescope, a double-pendulum interferometer, and a four channel liquid-He cooled infrared detector system. The heart of the instrument is the double pendulum interferometer, a modification of the classical Michelson set-up. The four-channel detector system allows the simultaneous coverage of the most important absorption bands of ozone-relevant molecules between 5.2 and 13.3 µm.
The analogue data is sampled on-board, mixed with the information of the other channels and the housekeeping data, and sent to ground via telemetry at a data rate of 250 kbit/s. An uplink connection of 1200 bit/s ensures full commandability of the instrument during flight.
On ground, the raw data is split up again and stored immediately in a data base. At the same time, housekeeping data and interferograms can be viewed and processed to allow on-line evaluation of measured data and of instrumental health.
Balloon launched on: 5/4/1990
Launch site: Centre de Lancement de Ballons CLBA, Aire Sur L'Adour, Landes, France
Balloon launched by: Centre National d'Etudes Spatiales (CNES)
Balloon manufacturer/size/composition: Zero Pressure Balloon
End of flight (L for landing time, W for last contact, otherwise termination time): 5/4/1990
Balloon flight duration (F: time at float only, otherwise total flight time in d:days / h:hours or m:minutes - ): 6 h
The balloon was launched by auxiliary balloon method on May 4, 1990 from the CNES base of Aire Sur L'Adour. The balloon reached a ceiling height of 38 km and the gondola stayed there for about 4 hhours before and 2 hours after sunrise.
This was the second scientific flight that involved an early version of the instrument known as MIPAS-B1. After a first technological flight performed in 1989 the spectral resolution, the spectral coverage and the sensitivity were improved and as a result during the second flight, the instrument yielded already more than 2000 atmosperic useful spectra.
If you consider that this website is interesting or useful, you can help to keep it running with just the equivalent of the price of a cup of coffee. Click on the button on the right for more information.