Purpose of the flight and payload description
This flight was part of an air sampling program started at the Institute of Space and Astronautical Science (ISAS) of Japan by a group led by Prof. T. Itoh in 1978. The objective of the program was to obtain stratospheric and tropospheric air samples above Japan first using a balloon-borne grab sampler and later a cryogenic sampler. After each flight those samples were distributed among collaborating scientists whom analyzed minor constituents of the stratospheric air using precise analyzers in their laboratories.
A schematic diagram of the cryogenic sampler is shown at left (click to enlarge). It consisted mainly of 12 stainless-steel-sample containers, a liquid helium dewar, a receiver, a transmitter, a control unit and batteries; all these components were housed in an aluminum chamber pressurized at 1 x 105 Pa. The volume of each sample container was about 760 ml and the inner wall of each container was electrically polished. A motor-driven metal-to-metal seal valve was attached to each sample container. The other end of the motor-driven valve was connected to the sample intake through a manifold. Air samples were introduced into the containers through a 5 m bellows tube, of which one end was located 4 m below the bottom of the sampler to avoid possible contamination from it.
All parts were made of stainless steel and connected to each other with metal gaskets. After assembly, the leakage of the system was confirmed to be minimal by a mass spectrometer leak detector, and all tubing including the sample containers were evacuated for longer than one month to make their insides clean. Prior to air sampling, a similar evacuation was made for about one week, and then the valves were closed and the sample containers were cooled to -269º C by filling the dewar with liquid helium. Total weight of the sampler was about 260 kg.
To minimize contamination from the balloon, the sampler was suspended about 120 meters below the balloon by a stainless steel wire and air samples were collected at assigned heights during the descent of the balloon. The maximum height of air samples collected ranged from 25.0 to 35.0 km, depending on the size of the balloon used. At each assigned height, the motor-driven valves were opened and closed using a telecommand. The amount of air samples collected was approximately 25 liters. To avoid intrusion of contaminated surface air into the sample containers, the stainless-steel bellows tube was also evacuated together with the containers and its evacuation was broken when the balloon ascended to the maximum height.
Once the balloon flight was completed the sampler detached from the balloon and descended under his own parachute. The sampler was recovered within an hour of the impact in the sea surface. Back in the laboratory, the sample containers were laid for about one month to ensure that respective components ofthe air samples were mixed well in the containers before proceeding to analyze them.
Details of the balloon flight
Balloon launched on: 9/1/1989
Launch site: Sanriku Balloon Center, Iwate, Japan
Balloon launched by: Institute of Space and Astronautical Science (ISAS)
Balloon manufacturer/size/composition: Zero Pressure Balloon model B30 30.000 m3
Flight identification number: B30-59
End of flight (L for landing time, W for last contact, otherwise termination time): 9/1/1989
Balloon flight duration (F: time at float only, otherwise total flight time in d:days / h:hours or m:minutes - ): 13 h 45 m
Landing site: In the Japan Sea
- Cryogenic Air Sampling Group homepage Tohoku University website
- Height Distributions of Methane Derived from Laser Heterodyne Observations Technical Report, Upper Atmosphere and Space Research Laboratory, Tohoku University (1993)
- Measurement of 14C Concentrations of Stratospheric CO2 by Accelerator Mass Spectrometry Radiocarbon, North America, Vol34 Nº 3, 1992
- Measurements of the stratospheric carbon dioxide concentration over Japan using a Balloon-borne cryogenic sampler Geophysical Research Letters, 22(10), 1229-1232
- Research on Balloon-borne Whole Air Sampling System for Studying Stratospheric Minor Constituents Thesis by Hideyuki Honda -In Japanese- (2001)
- Scientific observations with balloons Institute of Space and Astronautical Science Annual Handbook 1989 (In Japanese)
- Seasonal variations in 14C concentrations of stratospheric CO2 measured with accelerator mass spectrometry Nuclear Instruments and Methods in Physics Research Volume 92, Issues 1-4, 3 June 1994, Pages 413-416
- Variations of stratospheric trace gases measured using a balloon-borne cryogenic sampler Advances in Space Research, Vol 30, Issue 5, 2002, Pag 1349
After running StratoCat in an "advertising free" basis for 16 years, I've joined "Ko-Fi" to get funding for the research I do. If you find this website interesting or useful, you can help me to keep it up and running.