Purpose of the flight and payload description
The large area solid angle double scatter neutron telescope was first conceived in 1968 by the Physics Department and Institute of Geophysics and Planetary Physics of the University of California, Riverside. The objective of the experiment was to measure the flux, energy and direction of albedo neutrons from the earth and to search for solar neutrons.
A schematic cross section of the neutron telescope is shown at left along with a photograph of the assembled detector, showing the location of the anticoincidence sheets and light pipes without the gondola cover (click to enlarge). The aluminum tanks, S1, and S1, were filled with liquid scintillator NE 223 (decaline). The tanks were 100 cm x 50 cm x 15 cm. They were mounted coplanar with a separation of 100 cm center to center. Each tank was optically separated into eight identical cells and each cell was viewed by a 5 inch Amperex XP 1040 photomultiplier tube. For optimum light collection the cell walls were painted white and each photomultiplier tube face was in contact with the liquid scintillator. A reservoir tank with both positive and negative pressure relief valves was attached to each of the main tanks to allow for thermal expansion and contraction of the liquid scintillator.
Extraneous material which might attenuate either the incident or scattered neutrons was avoided wherever possible. The walls of the tanks and the optical dividers were as thin as feasible structurally and the photomultipliers were mounted along two sides of the tanks, rather than on the top or bottom. The thickness of the top and bottom of the tank was 0.32 cm. The resultant dimension of the tank-photomultiplier tube assembly was 102 cm x 110 cm x 16 cm.
The overall dimensions of the detector were limited by the requirements placed on energy response, resolution and efficiency. The efficiency of the detector was determined by the areas and thicknesses of S1 and S2 and the solid angle subtended at one tank by the other. With the additional constraint on up/down symmetry, the optimum detector thickness was about 15 cm for neutron energies between 2 and 100 MeV. If it were thicker, there would be some increase in efficiency but multiple neutron scattering would degrade the angle measurement.
Charged particles were rejected by completely enclosing each liquid scintillator tank with a box of Pilot F plastic scintillator, 6 mm thick. Each anticoincidence scintillator assembly had sheets 104 cm x 114 cm top and bottom, and four sheets interleaved to form the sides. The four side sheets were curved at the ends to produce overlap at the corners of the box. Each was viewed with a 2 inches photomultiplier tube (RCA 8575) via an adiabatic light guide. The corners of the "box" were rounded to reduce the maximum dimension of the detector. The top sheet of the anticoincidence scintillator box was viewed by two 5 inches photomultipliers (Amperex XP 1040) via adiabatic light guides, and the bottom sheet was viewed by two 2 inches photomultipliers (RCA 8575) via tapered light guides.
The liquid scintillator tanks were supported at the corners by 2.5 cm square cross section, 6 mm wall thickness chrominum-molybdenum steel tubing. These strong support members were required for the 360 Kg detector to survive the parachute landing following a balloon flight. The tubing penetrated the lower sheet of the top anticoincidence shield (A12) and the top sheet of the bottom anticoincidence shield (A21). It was cut to a "U" cross section where it penetrated the scintillator to reduce the amount of scintillator removal. The frame was attached to the base of the aluminum gondola. The gondola maintained the instrument at atmospheric pressure and near room temperature during a balloon flight. It had a 3.2 mm cylindrical wall and 1.6 mm spherical end caps and was surrounded by 15 cm of expanded polystyrene.
Details of the balloon flight
Balloon launched on: 5/14/1972 at 12:48 utc
Launch site: Columbia Scientific Balloon Facility, Palestine, Texas, US
Balloon launched by: National Center for Atmospheric Research (NCAR)
Balloon manufacturer/size/composition: Zero Pressure Balloon Winzen - 424.755 m3 (17.78 microns - 2 Caps. 17.78 microns Stratofilm)
Flight identification number: 676P
End of flight (L for landing time, W for last contact, otherwise termination time): 5/15/1972
Balloon flight duration (F: time at float only, otherwise total flight time in d:days / h:hours or m:minutes - ): F 30 h 15 m
Payload weight: 1164 kgs
The balloon was launched from NSBF balloon base in Palestine, Texas at 12:48 utc on May 14, 1972. After a nominal ascent the balloon reached a float altitude of 4 g/cm2 and had a flight duration of about 20 hours.
- Final Technical Report on Contract N00014-69-A-0200-5004 and N00014-76-C-0147 Report for 1 Apr 71-30 Nov 77 - University of California, Riverside
- National Scientific Balloon Facility Annual Report FY 1973 National Center for Atmospheric Research, September 1973
- Balloon Flight Record Nº 43, National Center for Atmospheric Research, Summer 1972 - StratoCat's private collection
- Upper limits to the quiet-time solar neutron flux from 10 to 100 MeV Astrophysical Journal, vol. 207, July 15, 1976, pt. 1, p. 630-638
If you consider that this website is interesting or useful, you can help to keep it running with just the equivalent of the price of a cup of coffee. Click on the button on the right for more information.